How to run a self-test on the 33120A Waveform Generator.
- Press and hold the [Shift] key while turning the power on.
- When when you hear the 2nd beep release the shift key.
- The test will run and give a pass or fail indication on the display.
- This test will not diagnose a faulty output fuse that has a symptom of no output. If this is the case, please use the instant message box to notify us of the station and room that has the problem.